Reply To: RAW REPORT
A further point of clarification: the two grub screws in the probe holder are a little misleading: typically one would expect the screw to bottom out and then require some tightening torque to hold a clamp load. That is NOT the case with these two screws at all. These two screws function more as a ‘retaining ring’ for the probe to not come loose (the probe spring preloads the probe against thetwo probes screws), so they setve merely as a stopping point (in the forward-most position) for the probe to rest on.
I’m tempted to draw a diagram to show it better-let me nnow if that is needed…
I hope that helps with your understanding of how it functions?